The ITAN screen tension meter has been calibrated to zero before leaving the factory.
Users do not need to adjust it by themselves.
The tension of the net is expressed in N / cm, which indicates how much the tensile
force of the mesh yarn per cm width is usually expressed in N-Newton or Kg-kg.
Newton shows the actual meaning of the tensiometer; the reading display method:
The first reading of the tensiometer reading is incorrect. The second reading of the
tensiometer is normal and the tensiometer should be flicked. Mesh next to it.
|Product Name:||Mechanical tensiometer|
|Measurement error:||± 0.5N/CM|
|Units Of Measurement:||N/cm|
|Accessories:||Standard calibration glass|
|Net Weight:||About 650 g|
|Total Weight:||About 1150g|
|Note Shockproof:||Put it on its side after use|
|Type of mesh||Ty pes of printing||Tension IN / CM|
|Ultra high tension mesh||Circuit board and measuring scale||16-20|
|High-strength mesh||Multi-color four-color
overprinting machine printing
|High precision ultrafine mesh||High-precision
multilayer circit printing
mesh standard monofilament
mesh standard monofilament
Features of TITAN wire mesh tensiometer:
TITAN tension meter is suitable for the measurement of various screen tension,
Pointer indicating tension, unit: N/CM. The tension at any place of the tensioned
cloth can be detected, the radial tension and weft tension of the measured screen
can be distinguished, make the screen to reach the same tension in the latitude and
longitude direction. TITAN tension meter is a precision instrument, it needs to be kept dry and idust-free, and need to pay attention to anti-fall and anti-shock.
- Zero adjustment disc
- 0 N / cm (zero point)
- Glass calibration block
- Induction device
Measurement Methods And Steps
Before using the tensiometer to calibrate the tension meter, place the tensiometer
on the front of the zero adjustment plate(Do not use too much effort),Turn the scale
lock counterclockwise by hand until the scale disc rotates.Rotate the dial until the dot
is pointing to the pointer, and then turn the scale lock clockwise to lock the scale.
(Note: 1. Scale in the zero when the long needle is not necessarily and the horizontal
line into 90 degrees, left partial right is normal.2. Daily use, such as zero adjustment
plate appears large gap or broken should be replaced.
The steel mesh or screen plate non-printed surface up on the platform.
The entire tension meter on the smooth side of the outer frame 15-20cm on the net
surface.With a finger to tap the stencil or wire mesh surface a few times, then the
pointer is the point of view of the current stencil or wire mesh on the actual tension
(Newton readings).Place the steel screen or screen version horizontally (PCB contact
or screen print side up),Place the tension meter in the steel mesh or screen version of
the four corners and center, check the point of the tension value, and the minimum
value fill (stencil or screen use resume).Check whether the measured value of steel
mesh or screen tension is within the allowable range , if the measured value is not
standard, the feedback engineer handles.
When testing the tension, the wire mesh must be suspended;
The tensiometer needs to be zeroed before use.
It is forbidden to add lubricant to any part.
Pay attention to shock. If the instrument is dropped, it may damage or reduce the
measurement accuracy and sensitivity.